Our Opening Hours Monday to Friday 8:30 am to 2:30 pm Address: 96 Ti Rakau Drive Pakuranga Auckland New Zealand Phone: 09 577 2616 Mobile: 0272215936 Fax: 09 355 0365 Email firstname.lastname@example.org
Free Prescriptions for Fully Subsidised Medicines, please upload your prescription during ordering products. Save Petrol Save Money & Save time, we are 100% NZ owned. Ask your doctor to send Email to email@example.com
Wash hands and fingertip sample site with soap and warm water. Rinse and dry thoroughly.
Unscrew and remove the lancing device tip.
Firmly insert a new lancet into the lancet holder. Hold the lancet firmly. Gently twist to pull off the protective disk. Save disk to recap lancet after use. Replace lancet device tip.
Turn the adjustable tip until it is aligned with the load confirmation window and release button.
Select a desired depth of one – to – seven (1-7) on the lancing device’s adjustable tip. Choose a depth by rotating the top portion of the adjustable tip until the setting number matches the arrow.
To cock the lancing device, hold the body of the lancing device in one hand. Pull the sliding barrel with the other hand. The device is loaded when you feel a click and the load confirmation window turns red.
Place the device against the pad of the finger. Press the release button. Wait a few seconds for a blood drop to form. Apply the blood sample to the narrow end of the test strip till the meter beeps. See instructions for CareSens N, or CareSens N POP meters and CareSens N Test Strips.